IMPORTANT SAFETY INFORMATION
Indications for Use
Dilapan-S® is for use by healthcare professionals trained in OB/GYN whenever cervical softening and dilation are desired, such as for cervical ripening during term labor induction or gynecological procedures that require cervical preparation.
Contraindication: Dilapan-S® is contraindicated in the presence of clinically apparent genital tract infection.
Warnings & Precautions:
- Dilapan-S® is intended for single use only. Do not reuse, resterilize, reprocess, or use if primary packaging has been opened or damaged. Discard after use.
- Careful placement of the device is essential to avoid traumatic injury to the cervix or uterus (see Instructions for Use—Insertion). The device should not be left in place more than 24 hours. Instruct patients to: Report any excessive bleeding, pain, or temperature elevation, and to avoid bathing, douching, and intercourse. Patients should return to the physician for removal of Dilapan-S® at the indicated time and should be instructed not to attempt self-removal under any circumstances.
- Potential Complications/Risks: Twisting of device during removal may cause the device to break (see Instructions for Use—Removal). Complications may include: device entrapment and/or fragmentation, expulsion, or retraction; patient discomfort or bleeding; spontaneous rupture of membranes; spontaneous onset of labor; cervical laceration.
Storage & Handling:
Store between +15°C and +30°C and keep away from direct sunlight and high humidity.
Please click here to download the full Instructions for Use.
You may report a product complaint or adverse event related to the
use of Dilapan-S® by calling 1 (888) 257-9676 (United States Only).
To contact Medicem directly, please email: email@example.com.